Cantilever 
Application 
Similar to 
Short specs
(typical values) 
A 
Contact Mode 
Contact 
Res. Frequency:
15 kHz 
Force Constant:
0.2 N/m 

Tipless AllInOne Probe, Cantilever A Specs 
Resonant Frequency 
Typical Values 
Range 
15 kHz 
+/ 5 kHz 
Force Constant 
0.2 N/m 
0.04 – 0.7 N/m 
Resonant Frequency and Force Constant Similar to 
Contact 
Cantilever Length 
500 μm 
+/ 10 μm 
Mean Width 
30 μm 
+/ 5 μm 
Thickness 
2.7 μm 
+/ 1 μm 
Coating 
none 

B 
Force modulation, Pulsed Force Mode 
Multi75 
Res. Frequency:
80 kHz 
Force Constant:
2.7 N/m 

Tipless AllInOne Probe, Cantilever B Specs 
Resonant Frequency 
Typical Values 
Range 
80 kHz 
+/ 30 kHz 
Force Constant 
2.7 N/m 
0.4  10 N/m 
Resonant Frequency and Force Constant Similar to 
Multi75 
Cantilever Length 
210 μm 
+/ 10 μm 
Mean Width 
30 μm 
+/ 5 μm 
Thickness 
2.7 μm 
+/ 1 μm 
Coating 
none 

C 
Soft Tapping Mode, Intermittent ContactMode 
Tap150G 
Res. Frequency:
150 kHz 
Force Constant:
7.4 N/m 

Tipless AllInOne Probe, Cantilever C Specs 
Resonant Frequency 
Typical Values 
Range 
150 kHz 
+/ 80 kHz 
Force Constant 
7.4 N/m 
1  29 N/m 
Resonant Frequency and Force Constant Similar to 
Tap150G 
Cantilever Length 
150 μm 
+/ 10 μm 
Mean Width 
30 μm 
+/ 5 μm 
Thickness 
2.7 μm 
+/ 1 μm 
Coating 
none 

D 
Tapping Mode, Intermittent ContactMode 
Tap300 
Res. Frequency:
350 kHz 
Force Constant:
40 N/m 

Tipless AllInOne Probe, Cantilever D Specs 
Resonant Frequency 
Typical Values 
Range 
350 kHz 
+/ 150 kHz 
Force Constant 
40 N/m 
7  160 N/m 
Resonant Frequency and Force Constant Similar to 
Tap300 
Cantilever Length 
100 μm 
+/ 10 μm 
Mean Width 
50 μm 
+/ 5 μm 
Thickness 
2.7 μm 
+/ 1 μm 
Coating 
none 
